On Friday October 18 we had a half night scheduled for LMIRCam science, followed by a half night for phase sensing engineering. The LMIRCam observations went smoothly and had good conditions.
For phase engineering, we saw significantly more phase variation with our revised loop algorithm (see last post). The phase variations appeared to be nearly +/- 180 degrees per frame at 2 microns. This suggests a vibration problem. We tried shutting off
- our own compressor, turning off
- the mirror ventilation
- the facility chiller
- turning down the AO gain.
None of these seemed to affect the phase variations. So, we took as much data we could. We acquired:
- 10 micron nulling data at 30 Hz (see plot from Denis Defrere below)
- 2 micron phase data at 500 Hz (data still being analyzed)
- 4 micron dispersed fringe data at 3 Hz (shows large pathlength motions)
As the plot shows there seems to be some vibration at 10 Hz, but 200 nm is about 1/10 of a wave at 2 microns, while the 2 micron data (the error signal for our control loop) seemed to be showing up to 500-1000 nm variations. For the moment, it's a mystery what's going on until we complete the data analysis.
Saturday through Monday observations are LMIRCam science, so we have a couple days to determine our next step for phase sensing. We will resume tests on Tuesday night.
/groups/lbti/search/index.rss?tag=hotlist/groups/lbti/search/?tag=hotWhat’s HotHotListHot!?tag=hot2/groups/lbti/sidebar/HotListguestGuest2015-07-06 19:14:01+00:002015-07-06 19:14:01updated3Added tag - hotguestGuest2015-07-06 19:09:13+00:002015-07-06 19:09:13addTag2First createdguestGuest2015-07-06 19:09:00+00:002015-07-06 19:09:00created1wiki2015-07-06T19:14:01+00:00groups/lbti/wiki/6bdffFalseExport Control Training/groups/lbti/wiki/6bdff/Export_Control_Training.htmlGuest3 updatesExport Control Training
To take the UA Export Control training please go to:
UA Export Control Training
and under Required Training Courses select the link below
...Falseguest2015-07-06T19:14:01+00:00phinzPhil Hinz2014-10-05 06:41:02+00:002014-10-05 06:41:02updated7Added tag - hotphinzPhil Hinz2014-10-05 06:40:59+00:002014-10-05 06:40:59addTag6phinzPhil Hinz2014-10-04 03:51:08+00:002014-10-04 03:51:08updated5phinzPhil Hinz2014-10-04 03:48:32+00:002014-10-04 03:48:32updated4phinzPhil Hinz2014-10-04 03:43:21+00:002014-10-04 03:43:21updated3phinzPhil Hinz2014-10-04 03:28:04+00:002014-10-04 03:28:04updated2First createdphinzPhil Hinz2014-10-04 03:21:14+00:002014-10-04 03:21:14created1weblog2014-10-05T06:41:02+00:00groups/lbti/weblog/74d8cFalseLBTI October Run: Night 2/groups/lbti/weblog/74d8c/LBTI_October_Run__Night_2.htmlPhil Hinz7 updatesLBTI October Run: Night 2
Night 2 of our run was clear and calm. We continued Kepler followup in the first half of the night. Justin Crepp and his team were here to obs...Falsephinz2014-10-05T06:41:02+00:00hot/groups/lbti/search/index.rss?sort=modifiedDate&kind=all&sortDirection=reverse&excludePages=wiki/welcomelist/groups/lbti/search/?sort=modifiedDate&kind=all&sortDirection=reverse&excludePages=wiki/welcomeRecent ChangesRecentChangesListUpdates?sort=modifiedDate&kind=all&sortDirection=reverse&excludePages=wiki/welcome0/groups/lbti/sidebar/RecentChangesListmodifiedDateallRecent ChangesRecentChangesListUpdateswiki/welcomeNo recent changes.reverse5searchlist/groups/lbti/calendar/Upcoming EventsUpcomingEventsListEvents1Getting events…